Used within the NHS ICU's to protect from Covid 19.
Up to a months average usage from one set of filters.
JSP recommends 1-3 days for high use, 3 - 5 days for medium use and up to 28 days for light use.
Durable thermoplastic rubber offering a superior fit to most face shapes.
Reflective strips create increased visibility in low-light environments for added safety.
Low resistance exhalation valve for easy breathing, with stable mask configuration.
Cost effective Force™ 8 filters available with low profile angle for minimum visual impairment.
With these filters you can instantly check that you have the correct seal time.
Put the straps and harness over your head and pull the straps to a suitably tight and comfortable fit.
Press the front and backs of both filter covers together to stop air from entering through the filters.
No air should come through the mask. Adjust the mask and repeat until it doesn't. Release when seal is made.
Features & benefits
Daily Face Fit Reassurance.
Protection against construction dust.
Inhalation valve for easy maintenance.
Low profile covers protect filters.
Cleaning advice for Covid19:
Reusable masks such as the Force™8 and Force™10 have been developed with cleaning as part of the design specification. However the cleaning methods used in the products normal application are not suitable for killing the Coronavirus.
Limited trials of different disinfection processes have been carried out. These show positive results when combined with the checks that are detailed in the standard user instructions for the products.
Wipes – Clinell and Isopropyl alcohol - leaving the filters in place on the masks and cleaning all surfaces with the wipes. Allow the mask to air dry. The Force™8 mask harness elastic straps may need to be removed for additional cleaning depending on local requirements.
Immersion – solutions of Milton™, Actichlor™, Hydrogen Peroxide (maximum 3%) - remove the filters from the mask and immerse in a solution diluted as advised by the manufacturer. After the required period, remove and rinse under clean water ensuring all traces of solution are removed. Allow the mask to air dry.
Vapour treatment Hydrogen peroxide – hydrogen peroxide concentrations above 3% will have a detrimental effect on the polymer of the faceseal of the Force™8 and Force™10 masks. This sterilisation process is not suitable for the masks.
MEIKO TopClean M respirator cleaning and disinfection machine – using
program 1 and drying at 50°C in a drying cabinet, disinfectant: EW 80
Mat (Dosing: 1 %) from EW 80 Systeme GmbH. rinse aid: GT 500 from
etol-Werk Eberhard Tripp GmbH & Co.
A solution called Tristel can also be used.